Commissioning and quality assurances of the Intrabeam Intra-Operative radiotherapy unit
Purpose: The authors report comprehensive commissioning and quality assurance (QA) procedures for Intrabeam, Intra-Operative radiotherapy (IORT) unit. The Intrabeam system miniature X-ray source is a 50 kV and 40 µA unit.
Methods: The authors’ tests include measurements of output, beam deflection, isotropy, kVp and mAs measurements, quality index, isodose, reproducibility, linearity, depth dose verification, and 3D dose distribution. IC ionization chamber and the UNIDOSE dosimeter were used for the output commissioning. Probe adjuster/ionization chamber holder (PAICH) was used to check the mechanical straightness of the probe. For radiation tests, NACP parallel plate chamber, Standard Imaging electrometer, 30 × 30 × 30 cm3 IAEA water phantom, solid water slabs, EDR-2 Films with RIT software, and ionization based survey meters were used. Unfors Xi platinum edition kVp meter was used to measure the kVp and mAs.
Results: In mechanical QA test, X-Beam position (-0.09 mm), Y-Beam position (0.01 mm), and radial position (0.11 mm) errors were within the tolerance level. Isotropy test with PDA, survey meter, ion chamber, and film measurements also produced results within the specifications. Output measurements with PAICH and external chamber measurements were matched. Beam quality, linearity, and reproducibility values were ascertained at 50KV and 40 µA and found to be within limits. Isodose, 3D dose distribution, transverse, and horizontal profiles showed the good isotropy of the source.
Conclusion: The authors’ methodology provides comprehensive commissioning and calibration procedures for the Intrabeam system.
Beatty J, Biggs PJ, Gall K, et al. A new miniature x-ray device for interstitial radiosurgery: dosimetry. Med Phys 1996; 23:53-62.
Schneider F, Clausen S, Thölking J, et al. A novel approach for superficial intraoperative radiotherapy (IORT) using a 50 kV X-ray source: a technical and case report. J Appl Clin Med Phys 2014; 15:4502.
Intrabeam System PRS 500 with XRS. User manual for the core system. Issue 3.0. Oberkochen, Germany. Carl Zeiss Medical Company; 2007.
Biggs DS, Thomson ES. Radiation properties of a miniature X-ray device for radiosurgery. Br J Radiol 1996; 69:544-7.
Armoogum KS, Parry JM, Souliman SK, et al. Functional intercomparison of intraoperative radiotherapy equipment - Photon Radiosurgery System. Radiat Oncol 2007; 2:11.
Pimpinella M, Mihailescu D, Guerra AS, Laitano RF. Dosimetric characteristics of electron beams produced by a mobile accelerator for IORT. Phys Med Biol 2007; 52:6197-214.
Mills MD, Fajardo LC, Wilson DL, et al. Commissioning of a mobile electron accelerator for intraoperative radiotherapy. J Appl Clin Med Phys 2001; 2:121-30.
Strigari L, Soriani A, Landoni V, et al. Radiation exposure of personnel during intraoperative radiotherapy (IORT): radiation protection aspects. J Exp Clin Cancer Res 2004; 23:489-94.
Veronesi U, Orecchia R, Luini A, et al. Full-dose intraoperative radiotherapy with electrons during breast-conserving surgery: experience with 590 cases. Ann Surg 2005; 242:101-6.
Beal K, McCormick B, Zelefsky MJ, et al. Single-fraction intraoperative radiotherapy for breast cancer: early cosmetic results. Int J Radiat Oncol Biol Phys 2007; 69:19-24.
Vaidya JS, Tobias JS, Baum M, et al. Intraoperative radiotherapy for breast cancer. Lancet Oncol 2004; 5:165–73.
Lam SCP, Xu Y, Ingram G, Chong LF. Dosimetric characteristics of INTRABEAM flat and surface applicators. Transl Cancer Res 2014; 3: 106-11.
Park CC, Bevan A, Podgorsak MB, Pouliot J, Yom SS. Electronic Brachytherapy. Evaluation Subcommittee of ASTRO’s Emerging Technology Committee; 2008.
European Commission in Radiation Protection 91. Criteria for Acceptability of Radiological (Including Radiotherapy) and Nuclear Medicine Installations. Luxembourg: Office for Official Publications of the European Communitie; ISBN 92-828-1140-9: 1997. Available from http://ec.europa.eu/energy/nuclear/radiation_protection/doc/publication/091_en.pdf
This work is licensed under a Creative Commons Attribution 3.0 License.
International Journal of Cancer Therapy and Oncology (ISSN 2330-4049)
© International Journal of Cancer Therapy and Oncology (IJCTO)
To make sure that you can receive messages from us, please add the 'ijcto.org' domain to your e-mail 'safe list'. If you do not receive e-mail in your 'inbox', check your 'bulk mail' or 'junk mail' folders.