Derivation of equations to define inflection point and its analysis in flattening filter free photon beams based on the principle of polynomial function
Purpose: The objective of this work is to (1) present a mechanism for calculating inflection points on profiles at various depths and field sizes, and (2) study the doses at the inflection points for various field sizes at depth of maximum dose (Dmax) for flattening filter free (FFF) photon beam profiles.
Methods: Graphical representation was done on percentage of dose versus inflection points. Also, using the polynomial function, the author formulated equations for calculating spot-on inflection point on the profiles for both the 6MV and 10 MV energies for different field sizes at various depths.
Results: In a 10 MV FFF radiation beam, the dose at inflection point of the profile decreases as the field size increases. However, in 6MV FFF radiation beam, the dose at the inflection point initially increases with an increase in the field size up to 10 ×10 cm2 and decreases after 10 ×10 cm2. The polynomial function was fitted for both the 6 MV and 10 MV FFF beams for all field sizes and depths.
Conclusion: Polynomial function is one of the easiest ways of identifying the inflection point in FFF beam for various field sizes and depths. Graphical representation of dose versus inflection point for both FFF energies was derived.
Rout BK, Muralidhar KR, Ali M, et al. Dosimetric study of RapidArc plans with flattened beam (FB) and flattening filter-free (FFF) beam for localized prostate cancer based on physical indices. Int J Cancer Ther Oncol 2014; 2:1-9.
Klein EE, Hanley J, Bayouth J, et al. Task Group 142, American Association of Physicists in Medicine. Task Group 142 report: quality assurance of medical accelerators. Med Phys 2009; 36:4197-212.
Cashmore J. The characterization of unflattened photon beams from a 6 MV linear accelerator. Phys Med Biol 2008; 53: 1933-46.
Kragl G, af Wetterstedt S, Knäusl B, et al. Dosimetric characteristics of 6 and 10MV unflattened photon beams. Radiother Oncol 2009; 93:141-6.
Pönisch F, Titt U, Vassiliev ON, et al. Properties of unflattened photon beams shaped by a multileaf collimator. Med Phys 2006; 33:1738-46.
Hrbacek J, Lang S, Klöck S. Commissioning of photon beams of a flattening filter-free linear accelerator and the accuracy of beam modeling using an anisotropic analytical algorithm. Int J Radiat Oncol Biol Phys 2011; 80:1228-37.
Georg D, Kragl G, Wetterstedt Sa, et al. Photon beam quality variations of a flattening filter free linear accelerator. Med Phys 2010; 37:49-53.
Xiong G, Rogers DW. Relationship between %dd (10)x and stopping-power ratios for flattening filter free accelerators: a Monte Carlo study. Med Phys 2008; 35:2104-9.
Sauer OA. Determination of the quality index (Q) for photon beams at arbitrary field sizes. Med Phys 2009; 36:4168-72.
Ceberg C, Johnsson S, Lind M, Knöös T. Prediction of stopping-power ratios in flattening-filter free beams. Med Phys 2010; 37:1164-8.
Wang Y, Khan MK, Ting JY, Easterling SB. Surface dose investigation of the flattening filter-free photon beams. Int J Radiat Oncol Biol Phys 2012; 83: e281-5.
Titt U, Vassiliev ON, Pönisch F, et al. Monte Carlo study of backscatter in a flattening filter free clinical accelerator. Med Phys 2006; 33:3270-3.
Mesbahi A. A Monte Carlo study on neutron and electron contamination of an unflattened 18-MV photon beam. Appl Radiat Isot 2009; 67:55-60.
Fogliata A, Garcia R, Knöös T, et al. Definition of parameters for quality assurance of flattening filter free (FFF) photon beams in radiation therapy. Med Phys 2012; 39: 6455-64.
Carpenter RG. Principles and procedures of statistics, with special reference to the biological sciences. The Eugenics Review 1960; 52:172-3.
This work is licensed under a Creative Commons Attribution 3.0 License.
International Journal of Cancer Therapy and Oncology (ISSN 2330-4049)
© International Journal of Cancer Therapy and Oncology (IJCTO)
To make sure that you can receive messages from us, please add the 'ijcto.org' domain to your e-mail 'safe list'. If you do not receive e-mail in your 'inbox', check your 'bulk mail' or 'junk mail' folders.